For further information, please visit the homepage of this product.
Conditions of Use
Single ICUMSA® Methods are only offered digitally. The subscription license includes two year access to the method, which can be read on sugarindustry.info.
The Method cannot be downloaded, shared or duplicated. Please note the Copyright, Licensing and Trademark Policy of ICUMSA© and Bartens.
Access to Method
Soon after the order and successful payment we will activate the access.
The method is used by sugar cane mills and may be used for cane payment.
The method is used to determine the polarimetric sucrose content, the dissolved solids content and the fibre content of prepared cane.
It is also suitable for determining these components in bagasse.
For polarimetric sucrose contents in the range of 10 °Z to 15 °Z, the absolute difference between two results under repeatability conditions should not exceed 0.4 °Z, the absolute difference between two results under reproducibility conditions should not exceed 0.5 °Z. For refractometric dry substance contents in the range of (13 to 17) g/100 g, the absolute difference between two results under repeatability conditions should not exceed 0.5 g/100 g, the absolute difference between two results under reproducibility conditions should not exceed 0.7 g/100 g.
Please click here to open ICUMSA Method GS5-1 (2022).